The model CCX Loadcell is heavy duty weighing applications, inclubing hoppers, tanks, silos and high capacity scales.

- Insensitive to side load
- Double convex type
- Robust high capacity design

#### Specifications

Specifications |
Accuracy |

Rated capacity(R.C) |
5tf, 10tf, 20tf, 30tf, 50tf, 100tf, 200tf, 300tf, 500tf |

Rated output(R.O) |
1.5 mV/V±0.5% |

Nonlinearity |
0.1% of R.O. |

Hysteresis |
0.1% of R.O. |

Repeatability |
0.05% of R.O. |

Creep(20min) |
0.05% of R.O./20min |

Zero balance |
1% R.O. |

Temperature effect, on rated output |
0.03% of LOAD/10℃ |

Temperature effect, on zero balance |
0.03% of R.O./10℃ |

Temperature range, compensated |
-10~60℃ |

Temperature range, safe |
-20~80℃ |

Terminal resistance input |
350 ±5Ω |

Terminal resistance output |
350 ±5Ω |

Insulation resistance |
2000MΩ |

Excitation recommended |
10V DC |

Safe overload |
150% R.C |

Cable length |
Φ7±, 4coreshied, 3m |

#### Dimensions Table

Capacity |
A |
B |
C |
D |
E |
F |
G |
H |
J |
K |
L |
M |
RI |
Weight(kg) |

5 tf (49.03KN) |
45 |
56 |
16.5 |
40 |
8 |
30 |
100 |
80 |
10.5 |
116 |
30 |
19 |
60 |
3 |

10 tf 98.07KN) |
88 |
120 |
24 |
90 |
15 |
30 |
140 |
100 |
18 |
180 |
30 |
19 |
140 |
7.8 |

20 tf (196.1KN) |
88 |
120 |
38 |
90 |
15 |
30 |
140 |
100 |
18 |
190 |
35 |
22 |
140 |
9.6 |

30 tf (294.2KN) |
93 |
140 |
38 |
110 |
15 |
30 |
200 |
150 |
23 |
220 |
40 |
25 |
140 |
20.2 |

50 tf (490.7KN) |
137 |
180 |
50 |
140 |
20 |
30 |
200 |
150 |
23 |
280 |
50 |
25 |
180 |
24.5 |

100 tf (980.7KN) |
166 |
240 |
70 |
200 |
20 |
30 |
220 |
160 |
27 |
356 |
58 |
32 |
200 |
42.7 |

200 tf (1.961MIN) |
216 |
260 |
95 |
210 |
25 |
30 |
280 |
220 |
39 |
396 |
68 |
40 |
300 |
86.4 |

300 tf (2.942MIN) |
216 |
340 |
120 |
290 |
25 |
30 |
300 |
220 |
45 |
492 |
76 |
50 |
600 |
150.0 |

500 tf (4.903MIN) |
265 |
390 |
155 |
330 |
30 |
30 |
380 |
280 |
50 |
568 |
89 |
65 |
800 |
253.1 |

★Specifications are subiect to change without notice.